Several have noticed the following behavior in the last few nights with the RCT:
I believe these are all symptoms of the encoder slippage and worm gear oscillation. I believe they needing cleaning/re-greasing. I’ve put in a service request with KPNO. I still think part of the PSF is mirror support related, but we wont know for sure until we eliminate the other sources of confusion.
Tonight I was able to zero point the telescope, and test the guider. It appears that the pointing tolerance for accurate guiding is about 15″, or ~20 pix from the absolute center of the chip. Guiding works better now, but it does not improve the PSF substantially.